pulse output vortex flow meter with battery power supply pulse output flow meter and flow metre features Without moving parts, simple structure, easy for installation maintenance The frequency and the actual measured fluid flow is linear, zero drift, the performance is very stable. High...Send Inquiry
pulse output vortex flow meter with battery power supply
pulse output flow meter and flow metre features
Without moving parts, simple structure, easy for installation maintenance
The frequency and the actual measured fluid flow is linear, zero drift, the performance is very stable.
High accuracy, ± 1.0%, ± 1.5%
Wide measurement range , in the range of Reynolds number of 2 × 104 ~ 7 × 106, up to 1:20, plug-turndown ratio of up to 1:25
Low pressure loss (approximately orifice flow meter 1/4 to 1/2), it belong to the energy-saving flow meter.
High temperature, medium temperature can be measured up to + 350 ℃
1.pulse output vortex flow meter
2.LUGB smart display vortex flow meter
3.Temperature and pressure compensation vortex flow meter
Flow meter installation form
Installation of flow meter
■ Making hole on pipes according to hole dimensions, and the hole position should meet the requirement of strait pipeline.
■ Input the whole set of flow meter which connecting the top flange to the pipeline hole.
■ Weld and fix the position on flange and pipeline.
■ Remove the flow meter, welding the flange according to requirement, and clean the protruding part inside of pipeline.
■ Install the sealing gasket which has the same size of pipe on inner flange, input the flow meter to flange, the flow direction of flow meter should be the same as fluid, then fasten the bolts.
Installation of platinum resistance and pressure transmitter
■ If the medium needs split temperature and pressure compensation(like steam, compressed air), then it should install temperature sensor and press transmitter.
■ The temperature should be installed below the flow meter 4-8D, making a 25mm hole on fixed position,welding the platinum resistance base vertically or inclined on the hole, mount the platinum resistance to the base, and make sure no leakage.
■ Pressure transmitter should be installed below flow meter 3-5D, the hole position should at the position where the end of elbow welding vertically on the hole, mount the valve on the other end of the elbow, install pressure transmitter on the top of valve, the both sides of valve should make sure no leakage. If measuring high temperature medium, the elbow should be filled with water in advance, in order not to damage the pressure transmitter because of high temperature.
pulse output flow meter/flow metre/battery flow meter transport
sample order:courier ex:DHL,FEDEX,UPS etc
Large order:sea or air service